Sending serial triggers from PsychoPy to ETG-4000 fNIRS
In the process of designing my latest experiment in PsychoPy I realised that setting up the serial port connection is not the most obvious thing to do. I wrote this tutorial so that others (and future me) won’t have to waste time reinventing the wheel.
After creating an initial version of my experiment (looping over a wav file) I tried to figure out how to send a relevant trigger to my fNIRS. Luckily, the PsychoPy tutorial has a section about using a serial port, but after reading this I still wasn’t sure how to use the code with my script. After a quick brainstorm with the lab technician we figured out that a simple script (see below) is indeed sending triggers to the fNIRS:
To better understand the arguments of the serial.Serial class please consult the pySerial documentation .
The argument (‘COM1’) is the name of the serial port used. The second argument is the Baud rate , it should be the same as the Baud rate used in the Parameter/External settings of the ETG-4000:
The last line of the script is sending the signal through the serial port. In this example, it is “A ” followed by python’s string literal for a Carriage Return . That was one of the strings expected by ETG-4000, i.e. it was on the list previously set up in Parameter/Stim Measurement :
The easiest way for me to test whether I was sending correct signals was to use the Communication Test in the External tab (see the second screenshot). Once the test is started, you can run the Python script to test whether the serial signals are coming through.
If the triggers work as expected, the code sections for serial triggers can be embedded in the experiment. It can be done via GUI or code editor. That’s where to add code using GUI:
The next step is adding the triggers for the beginning, each stimulus/block, and the end of recording.
Here is an example of an experiment using serial port triggers to delimit blocks of stimuli and individual stimuli.
Due to the sampling rate, I had to add delay between the triggers delimiting the blocks, otherwise they would not be captured accurately. The triggers for block sections had to be send consecutively because the triggers cannot be interspersed (not sure if that’s because of my settings). For instance, AABBAA is fine, but ABABAB is not.
本文开发（python）相关术语:python基础教程 python多线程 web开发工程师 软件开发工程师 软件开发流程
本文标题：Sending serial triggers from PsychoPy to ETG-4000 fNIRS